Re: Accuphase T-100
toggle quoted messageShow quoted text
The oxidized lubricant in the shaft bushings is the culprit. Get yourself some Kroil and apply to the shaft bushings. A little heat may help as well. This was my typical treatment for frozen potentiometers and of course all the frozen Teac pinch roller arm pivot bushings.
On Tue, Jan 11, 2022, 10:06 PM tune in <atpotter@...> wrote:
Picked this up today, it is in excellent cosmetic condition, and the alignment seemed good, despite sitting for the past 5 years according to the seller. Opened it up and looks like new, no oxidation or corrosion visible anywhere.