toggle quoted messageShow quoted text
Are your screen shots the delay in sending a DIT of the delay in
receiving a DIT?
Have you done any test in the delay between speaking in to the MIC and
hearing in the monitor?
- Paul KW7Y
At 02:58 PM 2/4/2020, you wrote:
On Feb 4, 2020, at 12:07
PM, Rob Sherwood <rob@...> wrote:
It takes about 5 minutes to make the measurement. Four minutes to hook up
the equipment and 1 minute to measure it on a digital scope.
I have data on 12 rigs at nc0b.com/latency
Download the PDF. I need to add the TD-890S.
On Feb 4, 2020, at 9:58 AM, Rob Sherwood <Rob@...>
ï»¿ That will be interesting with current rigs running from 10ms to
180ms. Rob, NC0B
On Feb 4, 2020, at 7:07 AM, Irwin Darack <idarack@...>
I realize that itâ€™s a little early and Elecraft is most likely still
fine tuning their firmware/software....has Elecraft published a expected
RX/TX Latency Time for the K4?
Groups.io Links: You receive all messages sent to this group.
View/Reply Online (#32048):
Mute This Topic:
Group Owner: Elecraft-K3firstname.lastname@example.org
Sent from my Android device with K-9 Mail. Please excuse my brevity.