toggle quoted messageShow quoted text
thanks for testing, but the issue that Akira san reported, is with BitX40,
If possible, could you please repeat the same check with BitX40 and let us
know the results?
Many Thanks, 73
On Sun, July 1, 2018 15:59, Raj vu2zap wrote:
Correction 500KHz span!
At 01-07-18, you wrote:
I just put my uBitxr4 on the spectrum analyzer and found nothing +-250
Khz. That is 500 Khz BW.
I think your measurements are being affected by switch mode power supply
in one of the items.
At 25-06-18, you wrote:
I built Bitx40 about 2 weeks ago.
I measured the spurious of Bitx40 by using the spectrum analyzer.
The level of spurious of harmonics were under 50db. So this is fine.
But the spurious at 75KHz was observed.
The spurious might be the multiple of 25KHz.
It would be much appreciated if you could advise me the cause and
solution of this spurious of Bitx40.
In Japan, we must prove to the authority that the spurious of the
transceiver meets the criteria to obtain the certification of the
transceiver from the authority.
So if i could not resolve this spurious, I will not be able to use
Bitx40 in Japan.
Thanks in advance,